Medizinische Universität Graz Austria/Österreich  Forschungsportal  Medical University of Graz
Gewählte Publikation:
SHR
Neuro
Krebs
Kardio
Lipid
Stoffw
Microb
Baulmann, J; Dorr, M; Genzel, E; Stauber, A; Richter, S; Ohlow, MA; Eckert, S.
Feasibility of Calculating Aortic Pulse Wave Velocity from Oscillometric Upper Arm Pulse Waves Using the Antares Algorithm
ARTERY RES. 2022; 28(1): 18.
Doi: 10.1007/s44200021000093
Web of Science
FullText
FullText_MUG
 Führende Autor*innen der Med Uni Graz

Baulmann Johannes
 Altmetrics:
 Dimensions Citations:
 Plum Analytics:
 Scite (citation analytics):
 Abstract:
 Background Antares is an algorithm for oscillometric blood pressure (BP) monitors to determine aortic pulse wave velocity (PWV) solely using oscillometric pulse waves without dependence of any other input. The aim of this study is to test Antares PWV for feasibility and whether the known age and blood pressure dependence of PWV can be shown with Antares PWV. Methods In total, 259 patients were investigated for PWV as substudy of the invasive validation of Antares algorithm, of which 219 entered analyses. Noninvasive PWV determination by Antares algorithm, integrated into an oscillometric BP monitor (custo screen 400) was compared to five different ePWV equations based on age, BP, or both. Additionally, in a subset of 27 patients, comparison of ARCSolver PWV algorithm (MobilOGraph) with Antares PWV and ePWV was conducted. Results Mean differences +/ SD between Antares PWV and (A) ePWV (based on age and systolic BP) was  0.05 +/ 1.06 m/s (Spearman's rank correlation coefficient, r(S) = 0.805); (B) ePWV (based on age) was  1.75 +/ 1.17 m/s (r(S) = 0.829); (C) ePWV (based on mean BP) was  1.35 +/ 1.24 m/s (r(S) = 0.763), and (D) ePWV (based on age and mean BP) was  1.64 +/ 1.22 m/s (r(S) = 0.810) and  1.69 +/ 1.18 m/s (r(S) = 0.802). Comparison of Antares PWV with ARCSolver PWV revealed a mean difference of  0.65 +/ 1.31 m/s (r(S) = 0.854). Conclusion The Antares algorithm confirmed its feasibility to use an oscillometric BP monitor as a singlepoint measurement device to calculate aortic PWV with acceptable comparability and high correlation to both estimated PWV and ARCSolver PWV. Antares achieves these results solely based on analysis of waveform features without requiring any secondary input, like BP or age.
 Find related publications in this database (Keywords)

Pulse wave velocity

Pulse wave analysis

Antares PWV

Custo screen 400

ARCSolver

MobilOGraph

Estimated PWV